This manual contains installation and operation procedures for the following Page 3 . Considerations for Safe Operation of Your Biodex Multi-Joint System. Records 16 – 44 This manual covers operation procedures for the following Biodex .. Biodex Multi-Joint System 3, Rev 2 or higher, with the proper hardware. Bi doex. User Guide the Biodex for the next instruction which will be to press start the Biodex for the next instruction The PC will load the System 3 y software.
|Published (Last):||8 September 2005|
|PDF File Size:||19.60 Mb|
|ePub File Size:||12.46 Mb|
|Price:||Free* [*Free Regsitration Required]|
A strong rotator cuff is especially important for a balanced and smooth movement ofthe upper extremity.
BIODEX SYSTEM 3 PRO MANUAL ( # 8 3 5 – 0 0 0 ) Pages 51 – – Text Version | FlipHTML5
Page 10 5 Troubleshooting Page. The passive mode may be used to assist this process, especially for older persons who use their joints less frequently and through smaller ranges of motion.
This system should be operated only by qualified personnel. These injuries may be treated by working either passively, isokinetically, or eccentrical- ly at submaximal levels. Secure Hip attachment strap. Limits should never be set at points that are beyond the safe maximum allowable range of motion for the individual subject.
The Systtem mode may be used isokinetically in the agonistic manuxl and then passively in the antagonistic direction or vice versa. When biode limits are set, the subject must exert at least one-tenth of the torque limit to keep the shaft moving.
The positioning chair offers degrees of rotation in the horizontal plane with detente settings at degree intervals. Direction of movementFigure 4.
To rotate the seat in either direction, boodex the Seat Rotation Handle toward the rear of the seat. Located on the Dynamometer Yoke Pivot Plate, theDynamometer Position Color Code helps the user to quickly position the dynamometer accord-ing to the pattern selected. Align Shaft red dot with R or L and reattach.
Setup Mode General Instructions 1. The ankle may be rehabilitated in an elevated position if edema is present. Read and understand More information.
The patient may move passively at first and, as tolerated, begin to assist the motion in each direction. Placing your hands or fingers between the dynamometer input shaft or attachment and the mechanical ROM stops may result in sgstem injury. Eccentric torque up to ft-lb Nm Accommodates the demands of today’s athletes and sports medicine researchers. To move bioedx dynamometer, press down on either foot pedal and slide the dynamometer to the desired location. These buttons provide the subject with the ability to instantaneously terminate exercise in any mode.
Setup and Positioning Starting Movement: ContractionThe Contraction buttons are active for isokinetic and isotonic modes only. For additional technical advice, service or educational information, contact Biodex personnel atthe following address: The isokinetic mode may be used bi-directionally wystem emphasize one particular muscle group, e.
BIODEX SYSTEM 3 PRO APPLICATION/OPERATION MANUAL – PDF
The exact amount of such physiological overflow is still debatable. Range of Motion A guide for you after spinal cord injury Spinal Cord Injury Rehabilitation Program This booklet has been written by the health care providers who provide care to people who sjstem a spinal.
The Isometric mode can be used very effectively to initiate contractions submaximally. Reactive Eccentric Mode The Reactive Eccentric mode for submaximal neuromuscular re-education in the early phases of rehabilitation The System 4 responds only to the patient’s force output.
Press the Boodex button to stopdynamometer shaft rotation cut off power. If you donot support the Cervical Support, it may slide down and pinch your hand as you loosen the knob. Manula quadriceps only, choose concentric away and eccentric towards.
Varying the number of repetitions i. Instruct subject to reposition and lie supine on chair. Turn the main power switch on manula back of the controller to the ON 1 position. This is especially important when dealing with fragile surgeries post-operatively. If needed raise chair or adjust seat back tilt to accommodate various patient sizes.
Low torque limits require greater neuromuscular control. Always assume that previously set limits are inappropriate for successive subjects, or for successive joints on the same subject. Install limb support angled away from patient in chair side receiving tube.
One 30 minutes period. Before starting any new exercise More information. Resistance continuously matches effort, accommodating to variations in patient force output due to weakness, pain or fatigue at specific points in the individual’s range of motion.
System 4 Software Update v4.60
These buttons provide the subject with the ability to instantaneously terminate exercise in any mode. Also displays system status codes in the case of malfunction. Provide neurologically impaired individuals with enough time to prepare for a contraction. Use the Pause Toward to increase or decrease the time length of the pause in the toward direction. Always press the Setup button on the control panel prior to setup of a new subject or before subsequent setups on the same subject.
Footrest optional Positioning Chair: The Isotonic mode can be utilized to ensure a proper force limit is achieved.
To tilt the dynamometer, supportthe dynamometer with one hand. The passive mode may be used for continuous passive motion effects. To challenge scapula thoracic stability with dissociated movement. The pause may also be used when working in the passive mode to do eccentric or non- reciprocal contractions.
Concentric torque up to ft-lb Nm and eccentric torque up to ft-lb Nm accommodates the demands of high performance athletes and sports medicine researchers. Ensure mabual all system wiring and cables are routed away from any area where they might sysfem stepped on or rolled over by wheeled equipment.